Wolfgang Renz1,2, Tomasz Lindel, 23, Matthias Dieringer2,4, Frank Seifert, 23, Jeanette Schulz-Menger, 2,5, Thoralf Niendorf2,4, Bernd Ittermann, 23
1Siemens AG, Erlangen, Germany; 2Berlin Ultrahigh Field Facility, Max-Delbrueck-Center for Molecular Medicine, Berlin, Germany; 3Physikalisch-Technische Bundesanstalt (PTB), Berlin, Germany; 4Experimental and Clinical Research Center, Charite Campus Buch, Berlin, Germany; 5CMR-Unit, Charite Campus Buch, Berlin, Germany
The first steps in cardiac/body imaging at 7T have been reported. One of the challenges is a suitable TX coil concept, which adresses the RF problems (B1 homogeneity, SAR) of body imaging at 300MHz. Traditional bodycoils seem not to be the right way, a coil array enabling TX SENSE and B1 shimming seems to be more promising. We describe a 8-element TX/RX loop coil array with adjustable capacitive decoupling. A prototype has been realised and tested. First imaging results in cardiac imaging are shown.